High-voltage cables are an essential part of X-ray machines, DR, CT and other equipment. Once oil leakage or ignition is found, they must be stopped immediately and repaired and replaced. Otherwise, equipment damage or safety accidents may be caused, resulting in heavy losses. Be sure to purchase a new cable that meets the voltage requirements before replacing the high-voltage cable. If the tube is 125kv, the high-voltage cable can be used with 75kv. If the tube is 150kv, the high-voltage cable can be replaced with 90kv. Before purchasing, be sure to check the voltage of your equipment, the length of the cable, whether the plug is a national standard plug, the number of wire cores, etc. Otherwise, it may not match and cannot be used. When receiving a new cable, be sure to check whether the length is correct, whether the plug is the same size and specification, whether the cable sheath is damaged or broken, etc., apply insulating silicone grease to the high-voltage plug and insert it slowly into the socket of the ball tube, and make sure the contacts of the plug are aligned If there is a hole in the socket, otherwise it cannot be installed in place. After the plug is inserted, the pressure ring screw can be tightened. There are two screws at the bottom of the pressure ring that can be used to fix the pressure ring to prevent the pressure ring from moving back and forth, causing the pressure ring to loosen and cause arcing of the plug. In the same way, the other end can be inserted into the high-voltage generator, and the power-on test is normal and available.
If you are interested in our high-voltage cables or other X-ray equipment, please contact us.Whatsapp:+86 18953613955.