Medical power cable is required for any X-ray machine
The medical power cable on the X-ray machine is connected to the bulb and generator.The high pressure generated by the high pressure generator is transported to the bulb, and the X-ray machine includes CR and DR systems.The rated temperature of medical power cable is 70 degrees, and the material is tinned copper.The core is red and white.In normal times, medical power cable should be kept dry and clean to avoid the erosion of oil, water and harmful gas, thus avoiding rubber aging.
Newheek Medical Power Cable Technical Requirement
Medical power cable sheath appearance should be smooth, uniform diameter, with no joint, bubble, bumps and other undesirable phenomena.
Weave shield density is not less than 90%.
The minimum thickness of the cable insulation and sheath must be 85% more than the nominal thickness.
Insulation between the core and the insulated wire, insulation between the core and the ground cable should be able to withstand the AC 1.5KV and keep 10 minutes can not been broken down.
Insulation between the core and the shield should be able to withstand the DC 90 KV and keep 15 minutes can not been broken down.
The DC resistance of the conductor and ground cable not more than 11.4 + 5%Ω/m.
The insulation resistance of the insulation core wire not less than 1000MΩ·Km.
The plug body should be able to withstand not less than 1000 times fallen off experiments with no damage.
The surface of each plating should be clean and bright.
The medical power cable and each part should meet Rohs 3.0 relative requirement. Brass is below 0.1wt.
The medical power cable and each part should meet Reach relative requirement.