High-voltage cable breakdown can be repaired. X-ray machine high-voltage cable breakdown mostly occurs near the high-voltage cable plug.
How to repair the high voltage cable after breakdown? First find the location of the breakdown. Generally, the breakdown location is blacked out. The second maintenance process is: after confirming and marking the connection position of the large and small focus and the common line, welding the metal shielding layer and the pin lead. Use heated transformer oil to melt the original filler in the plug and pull out the plug. Saw the breakdown part of the cable, and trim the protective layer and metal shielding layer of the high-voltage cable according to the length of the plug. Finally, encapsulate the plug.