The high-voltage cable connects the x-ray tube and the high-voltage generator. There are 75KV and 90KV high voltage cables. As the connection part, the high-voltage cable has the risk of being damaged and broken down, leakage, and oil leakage. The failure of the high-voltage cable (90kv high-voltage cable) of the X-ray machine usually occurs near the plug of the high-voltage cable. How to repair the fault of the high-voltage cable, first, find the location of the fault (90kv high-voltage cable). Usually, when the fault location is black, the fault location will be found. Secondly, the maintenance process is: confirm and mark the wiring position of the focal point and the common line, and then solder the metal shielding layer and the pin leads. Melt the original filler in the plug with heated transformer oil, and then pull out the plug. Saw off the breakdown part of the cable, and trim the protective layer and metal shielding layer of the cable according to the length of the plug. Finally, wrap the plug with insulating tape. Our company produces and sells various x-ray machine parts. This product is widely used in domestic and foreign medical and health fields. If you are interested in our products, please contact us. People from all walks of life are welcome to inquire and discuss supporting production, maintenance and replacement, agency trade and other matters.