The high voltage cable connects the X-ray tube and the high-voltage generator. There are 75KV and 90KV cables. As a connecting part, high voltage cable has the risk of being broken down,
The breakdown of high voltage cable of X-ray machine usually occurs near the plug of high-voltage cable.
How to repair the breakdown of high voltage cable,
First of all, find the location of the breakdown. In general, the breakdown position is found when the breakdown position is black. Secondly, the maintenance process is: confirm and mark the wiring position of the focal point and common line, and then weld the metal shielding layer and pin lead. Melt the original filler in the plug with heated transformer oil and pull out the plug. Saw off the breakdown part of the cable, and trim the protective layer and metal shielding layer of the cable according to the length of the plug. Finally, pack the plug.
Our company Shandong newheek produces and sells all kinds of X-ray machines and parts. The products are widely used in medical and health fields at home and abroad. If you are interested in our products, please contact us.